|Home||» Products||» Rapid Test Kits||» Mumps Virus IgM rapid test cassette|
NAME AND INTENDED USEThis rapid test for IgM Antibody to Mumps virus (Colloidal Gold) is an in vitro immunoassay for the qualitative detection of the mumps virus IgM antibodies present in the human whole blood, serum or plasma samples, and is used as an auxiliary diagnostic reagent for the early infection of mumps virus in clinical diagnosis.
Mumps Virus (Mu) is a pathogen of mumps and belongs to the Paramyxoviridae family. It was first discovered in 1935 and was proved to be the main cause of mumps. This virus is on its peak in winter and spring and is limited to humans. The main routes for the transmission of this virus are saliva, saliva-contaminated foods and articles, and airborne droplets. Children and adolescents are more susceptible to mumps. Parotid gland enlargement is the main characteristic, and common complications are meningitis and encephalitis, epididymis inflammation, orchitis, ovarian inflammation, pancreatitis, myocarditis. Mumps virus IgM antibody appear shortly after the onset between 2 to 3 days and reach its peak within 2 weeks, and then it decreases gradually that could be for several months. At present, the main methods to diagnose this MuV infection in clinic or laboratory includes virus isolation and culture, serological detection (complement binding test, neutralization test, etc), immunological detection (enzyme-linked immunoassay, indirect immune-fluorescence detection, etc) and molecular biological methods such as RT-PCR.
PRINCIPLES OF THE ASSAY
This test kit uses a colloidal gold immunochromatography assay, the colloidal gold-labeled mouse-anti-human IgM is coated on the nitrocellulose membrane. If the sample is a positive specimen then the mumps virus IgM antibody in the specimen is combined with the colloidal gold-labeled-mouse. The anti-human IgM binds to form an immune complex, and the complex with the sample flow forward inside the nitrocellulose membrane due to chromatography. When the complex passes through the T line, it binds with the encapsulated mumps virus antigen to form the complex and agglomerate and display a colored line. The remaining colloidal-gold labeled mouse-anti-human IgM and the anti-mouse IgG antibody in the C-line is bound and agglutinate for color development. If the sample is negative, means the sample does not contain mumps virus IgM antibody, resulting in the inability to form an immune complex and color only appears only in the C line.
20 x1 tests (Cassette) Mumps Virus Test Cassette
Each box contains as below:
Test card (T-line coated with recombinant mumps virus antigen (NP), coated with goat-anti-mouse IgG antibody on C line, coating buffer: 0.05 M PBS pH 7.4): 1 test× 20
Sample diluents: 1 bottle × 5 ml.
Manual: 1 copy
The components in different batches and of different varieties of kits should not be mixed.
Serum samples are collected from the intravenous method by the conventional method. Plasma sample: add 100ul heparin solution (1%) to 5 to 10ml of blood; or sodium citrate solution (3.8%) to plasma according to the proportion of 1:9; or EDTA solution (15%) 0.04ml to 5ml plasma.
Serum and plasma samples can be stored at 4℃ if test is done within 5 days or else store for more than 5 days store at -20 ℃ and not more than 3 times of freeze-thaw.
Whole blood samples are recommended to be tested within 3 days and should be stored at 2 to 8 °C, do not freeze.
The test result is invalid for hemolytic sample.
Test preparation: 1μL, of each of 10μL, 50μL and 100μL micropipettes, and matched tips or dropper (10μL, 50μL, 100μL)
Test process: Make sure that before the test the temperature of the kit and the sample should be the same with the room temperature. Place the test card on a horizontal dry work surface. Add 10μL serum or plasma sample into the sample well then add 20μL of whole blood to each well of the test card using a micropipette, add 80μL of the sample dilution. Observe the result in 15 to 20 minutes. The observation is invalid after 20 minutes.
Precautions taken during the inspection process: The test kit and the sample to be tested should be placed at room temperature before being tested. Use the test card within 30 minutes after opening it.
INTERPRETATION OF RESULTS
Positive: If two pink lines appear in both the C and T position
Negative: If only one pink line appears in the C position
Invalid: If the control line next to the test line is not visible within 10 minutes after the addition of the sample.
Note: Invalid result should be treated as infectious contaminants and samples should be re-acquired.
The test results of this product are interpreted by human and are subjected to visual errors or subjective judgment. Therefore, when the color of the strip is difficult to determine it clearly, it is recommended to repeat the test.
This test card is one of the auxiliary means of diagnosis and the test result is for reference only so, should not be taken as the sole basis for clinical diagnosis and treatment. The positive result should be further confirmed by other methods also, the detection result is limited by the sensitivity of the test. The negative result may be due to low antibody concentration, due to the sensitivity of product analysis. Clinical diagnosis should be combined with clinical examination, medical history and other tests.
In the early stage of infection, IgM is not produced or the titer is very low that may lead to negative result. Patients should be reminded to review within 7 to 14 days. At the same time, the last collected specimens should be tested in parallel to confirm whether there is serological positive or significant titer rise.
Patients with impaired immune function or immunosuppressive therapy have limited reference values for serological antibody testing.
This reagent is a qualitative test and cannot be used to determine the antibody content.
This reagent is for the detection of individual whole blood, serum or plasma samples do not use it for the detection of saliva, urine or other body fluids.
The positive result detected by the kit further needed to be confirmed by other methods.
The kit should be sealed and protected from moisture when the humidity is 60% or less, should be used within 1 hour of opening and if the humidity is more than 60%, avoid keeping in the air for too long, that can cause moisture absorption and affect the test results.
The depth of the color of the test line is not necessarily related to the titer of the antibody in the sample. The result after 20 minutes is invalid.
When the mumps virus IgM antibody content in the specimen is extremely high, the C-line band may be weak, which is normal.
The kit components and the waste generated by the test should be treated as infectious contaminants.
This reagent is for in vitro diagnosis only.
STORAGE AND STABILITY
Store at room temperature between 4 to 30° C and the shelf life is of 24months. Do not use the kit beyond its expiry date.